|Plate Reading Velocity:||5s Of Single Waveform, 10s Of Dual Wavelength;||Accurate Wavelength:||±3.0nm|
|Warm Up Time::||10 Minute|
Portable Elisa Microplate Reader,
Large LCD Display Elisa Microplate Reader,
Elisa Reader And Washer
Before you start installing and working with the reader, you should read the safety precautions and regulations shown in this chapter.
Please note that the operation with Microplate Reader should be carried out only by the doctor or clinical inspector who has undergone necessary training provided by the sales agent.
To install, maintain and repair the instrument, a service technician has to be trained on the instrument by the manufacturer or their representative. A service technician is also expected to be familiar with the normal operation of the instrument as described in the User’s manual and the special operations as described in the service manual.
To use analyser safely, pay attention to the following items:
To prevent the risk of electrical shock and/or damage to the instrument, Operator should not open the cover of the instrument. Only authorized personnel, for example, service technicians, may open the instrument to perform maintenance or repair.
Touching the main board when the power is on may cause severe injury or death. Any problem, please ask for helps from your supplier.
There is no risk presented by the mechanical parts of the instrument when the instrument is closed. If the covers of the instruments are removed, mechanical parts could cause personal injury or the instrument may be damaged if the following advice is not being considered: DO NOT wear loose garments or jewellery that could catch in mechanisms. DO NOT put your fingers/hands into the pathway or any part while the instrument is in operation. DO NOT attempt mechanical repair unless the instrument is not in operation or OFF.
The source lamp becomes extremely hot during operation; never touch the lamp when it is on! If the lamp needs to be changed, always switch off the lamp by switching off the instrument and then wait until the lamp has cooled down.
|Work Environment:||temperature 10℃-30℃；Humidity ≤70%|
|Store Environment:||-20℃-55℃；Humidity ≤93%|
|Standard Wavelength:||405,450,492,630nm,Ten optical fibers can be configured within 400 to 800 at a time;|
|Instrument sensitivity:||no less than 0.01L/mg.|