|Type:||Blood Testing Equipments||Temperature:||4~30℃|
A/B Antigen Combo Rapid Test Kit,
Influenza Combo Rapid Test Kit,
COVID-19 Rapid Test Kit
REAGEN This kit is used for in vitro qualitative detection of COVID-19 & Influenza A/B antigen in human oropharyngeal swabs and nasopharyngeal swab samples.
REAGEN This kit adopts the detection principle of double antibody sandwich method. The binding pad of the detection card contains colloidal gold-labeled mouse N protein monoclonal antibody I; the detection line (T&A/B line) of the nitrocellulose membrane is coated with mouse N protein monoclonal antibody Antibody II, quality control line (line C) is coated with goat anti-mouse antibody.
The collected samples are processed by the sample extract and added. During the test, when the extract containing the N protein of the COVID-19 & Influenza A/B is added to the sample hole of the test card, under the action of chromatography, the sample moves to the end of the absorbent paper and first passes through the binding pad. The N protein and colloidal gold labeled mice The source N protein monoclonal antibody I specifically binds and continues to move to the absorbent paper end. When the sample moves to the T&A/B line, the N protein bound to the labeled antibody specifically binds to the N protein monoclonal antibody II coated in the T&A/B line in a double antibody sandwich mode, and stays at the T&A/B line; the remaining liquid continues to absorb water The paper end moves. When it moves to the C line, the mouse N protein antibody I labeled with colloidal gold specifically binds to the goat anti-mouse antibody in the C line. The T&A/B line shows a red band, indicating that the COVID-19 & Influenza A/B antigen is positive. No matter whether the T&A/B line is colored or not, the C line should show red. If the C line is not colored, the test is invalid and the sample needs to be tested again.
Negative: A purple-red band, only the quality control line (C line) is colored.
Positive: Two purple-red strips, the detection line (T&A/B) and the quality control line (C line) are both colored .
REAGEN The sensitivity of the test kit was 95.5%, the specificity was 98.3%, and the accuracy was between 90%-99.0%.